With respect to the suffering and the condition of a patient, device selection often performs a decisive role in the possible therapeutic success. With appropriate devices selection you employ a high probability of success with an iontophoresis gadget to dry. Not necessarily with treatment strictly regarding to manual a success (independence from sweat) can be reached. In this case, we are of course and especially following the purchase of course – with this experience since 1999 in the field of iontophoresis – aside. With the method Iontophoresis, many have the ability to “get dry”.
really great feeling when hands / ft / armpits are dried out and warm rather than wet and cold (frog hands). Still, many users delivered their best iontophoresis machine back – why? We also asked ourselves this question, most importantly because we are also thinking about providing our customers with ideal and successful therapy.
Background details: There are two types of treatment streams: Pulse and DC. DC devices deliver the result of greater efficiency (sweat level of resistance) with the same current setting. Now you could come to the conclusion: Fine, I’m sweating heavily, so DC is better for me, and cheaper it is! Unfortunately this is simply not always correct: the therapy with pulse current products is merely more pleasant with the same current environment. Consequently, with a pulse-current device, you can often drive higher currents when compared to direct current, since the current is better tolerated.
However, the limit values must not be exceeded (Dr. Hoenle specifies no more than 25 mA for your feet, 15 mA for the hands and 5 mA for the treatment of the underarms as a limit). These values are not normally accomplished with immediate current products, unless the individual is completely insensitive to the treatment stream, which is very rare, nevertheless, and the skin must join the whole. Even on your feet, 30 mA DC is a word! If you can arranged 5 mA with a pulse power device under the armpits, this may be more than enough to record a success because the skin there is quite thin (and regrettably sensitive) and for that reason does not need higher currents. Specifically with the armpits, DC devices will often tolerate just 1mA and thus in some instances no therapeutic achievement may be accomplished. The same pertains to the hands, where you often have minor injuries (nail, etc.) where not merely direct currents can be extremely painful. It isn’t so much this issue of your toes, but also the rule of the term: The pulse current can be often a lot more pleasant there aswell.